Home › Forums › Legacy SDKs › Tobii T60 XL accuracy problems
- This topic has 3 replies, 3 voices, and was last updated 3 years ago by Grant [Tobii].
- 01/05/2016 at 17:08 #5125AnonymousInactive
Hi, I am using tobii T60 xl for infant behavioral studies. I am wondering how to improve the tracking accuracy. In particular, I am frequently having troubles in that participants’ eye-gazes are below AOIs.
Thanks in advance!02/05/2016 at 14:03 #5131Grant [Tobii]Keymaster
This forum is exclusively for the Tobii EyeX Tracker. Please contact Tobii Pro support for issues relating to the Tobii T60XL using their support form @ http://www.tobiipro.com/contact/contact-support/13/03/2020 at 12:55 #14031Fred UdoyeParticipant
Hi Tobii team,
I am using Tobii T60-XL eyetracker for my study. I presented participants some videos on the screen. Please is there a way I can define areas of interest (AOI) after they’ve finished watching the video? For instance, I want to analyse only fixations made within the AOI. Please any help with this would be most appreciated.
Fred15/03/2020 at 22:41 #14037Grant [Tobii]Keymaster
Hi @fred4, as state above, for support on the T60-XL which is part of the Tobii Pro range of eye tracking hardware, please get in touch with their dedicated support team @ http://www.tobiipro.com/contact/contact-support/ for an answer to your query.
Apologies for any inconvenience and thanks for your understanding.
- You must be logged in to reply to this topic.